- AnesthesiologyReview Panel
- CardiovascularReview Panel
- ChemistryReview Panel
- DentalReview Panel
- Ear, Nose, ThroatReview Panel
- Gastroenterology and UrologyReview Panel
- General HospitalReview Panel
- HematologyReview Panel
- ImmunologyReview Panel
- MicrobiologyReview Panel
- NeurologyReview Panel
- Cardiovascular Diagnostic DevicesCFR Sub-Part
- Cardiovascular Monitoring DevicesCFR Sub-Part
- Cardiovascular Therapeutic DevicesCFR Sub-Part
- Diagnostic DevicesCFR Sub-Part
- Neurological Diagnostic DevicesCFR Sub-Part
- GXWEchoencephalograph2Product Code
- GXXBlock, Test, Ultrasonic Scanner Calibration1Product Code
- GXYElectrode, Cutaneous2Product Code
- GXZElectrode, Needle2Product Code
- GYATester, Electrode/Lead, Electroencephalograph1Product Code
- GXSMonitor, Alpha2Product Code
- GWEStimulator, Photic, Evoked Response2Product Code
- GWFStimulator, Electrical, Evoked Response2Product Code
- GWGEndoscope, Neurological2Product Code
- GWIDiscriminator, Two-Point1Product Code
- GWJStimulator, Auditory, Evoked Response2Product Code
- GWKConditioner, Signal, Physiological2Product Code
- GWLAmplifier, Physiological Signal2Product Code
- GWMDevice, Monitoring, Intracranial Pressure2Product Code
- GWNNystagmograph2Product Code
- GWQFull-Montage Standard Electroencephalograph2Product Code
- GWRGenerator, Electroencephalograph Test Signal1Product Code
- GWSAnalyzer, Spectrum, Electroencephalogram Signal1Product Code
- GWWAtaxiagraph1Product Code
- GWXFork, Tuning1Product Code
- GWYPinwheel1Product Code
- GWZPercussor1Product Code
- GXBEsthesiometer1Product Code
- GYBMedia, Electroconductive2Product Code
- GYCElectrode, Cortical2Product Code
- GYDTransducer, Tremor2Product Code
- GYEEncephalogram Telemetry System2Product Code
- GZKElectrode, Nasopharyngeal2Product Code
- GZLElectrode, Depth2Product Code
- GZMAnalyzer, Rigidity2Product Code
- GZNRheoencephalograph3Product Code
- GZODevice, Galvanic Skin Response Measurement2Product Code
- GZPStimulator, Mechanical, Evoked Response2Product Code
- HCJDevice, Skin Potential Measurement2Product Code
- HCSDevice, Temperature Measurement, Direct Contact, Powered2Product Code
- JXEDevice, Nerve Conduction Velocity Measurement2Product Code
- JXFPlethysmograph, Ocular3Product Code
- LLNDevice, Vibration Threshold Measurement1Product Code
- LQWTest, Temperature Discrimination1Product Code
- NCGNeuropsychiatric Interpretative Electroencephalograph Assessment Aid2Product Code
- NTUEvoked Potential Stimulator, Thermal2Product Code
- OLTNon-Normalizing Quantitative Electroencephalograph Software2Product Code
- OLUNormalizing Quantitative Electroencephalograph Software2Product Code
- OLVStandard Polysomnograph With Electroencephalograph2Product Code
- OLWIndex-Generating Electroencephalograph Software2Product Code
- OLXSource Localization Software For Electroencephalograph Or Magnetoencephalograph2Product Code
- OLYMagnetoencephalograph2Product Code
- OLZAutomatic Event Detection Software For Polysomnograph With Electroencephalograph2Product Code
- OMAAmplitude-Integrated Electroencephalograph2Product Code
- OMBAutomatic Event Detection Software For Full-Montage Electroencephalograph2Product Code
- OMCReduced- Montage Standard Electroencephalograph2Product Code
- ORTBurst Suppression Detection Software For Electroencephalograph2Product Code
- PIWBrain Injury Adjunctive Interpretive Electroencephalograph Assessment Aid2Product Code
- PKQComputerized Cognitive Assessment Aid2Product Code
- PNAEvoked Photon Image Capture Device1Product Code
- POMComputerized Cognitive Assessment Aid For Concussion2Product Code
- POPCranial Motion Measurement Device2Product Code
- POSPhysiological Signal Based Seizure Monitoring System2Product Code
- PSPIntracranial Pressure & Temperature Monitoring Kit2Product Code
- PTBAtaxiagraph With Interpretive Software1Product Code
- PTYComputerized Cognitive Assessment Aid, Exempt2Product Code
- PVPNeurological Test Kit1Product Code
- QEABrain Injury Adjunctive Interpretive Oculomotor Assessment Aid2Product Code
- OPTInfrared Hematoma Detector2Product Code
- Neurological Surgical DevicesCFR Sub-Part
- Neurological Therapeutic DevicesCFR Sub-Part
- Physical Medicine Diagnostic DevicesCFR Sub-Part
- Physical Medicine Prosthetic DevicesCFR Sub-Part
- Physical Medicine Therapeutic DevicesCFR Sub-Part
- Surgical DevicesCFR Sub-Part
- Therapeutic DevicesCFR Sub-Part
- Obstetrical and GynecologicalReview Panel
- OphthalmicReview Panel
- OrthopedicReview Panel
- PathologyReview Panel
- Physical MedicineReview Panel
- RadiologyReview Panel
- General and Plastic SurgeryReview Panel
- ToxicologyReview Panel
- Medical GeneticsReview Panel
- UnknownReview Panel
Neuropsychiatric Interpretative Electroencephalograph Assessment Aid
- Page Type
- Product Code
- Uses a patient's electroencephalograph (EEG) to provide an interpretation of the patient's neuropsychiatric condition. It is also used only as an assessment aid for a medical condition for which there exists other valid methods of diagnosis."
- Physical State
- Technical Method
- Software analysis of EEG signals
- Target Area
- Regulation Medical Specialty
- Review Panel
- Submission Type
- Device Classification
- Class 2
- Regulation Number
- GMP Exempt?
- Summary Malfunction Reporting
- Implanted Device
- Life-Sustain/Support Device
- Third Party Review
- Not Third Party Eligible
- MAUDE Alerts
- View and Sign Up For MAUDE Alerts
CFR § 882.1440 Neuropsychiatric interpretive electroencephalograph assessment aid
§ 882.1440 Neuropsychiatric interpretive electroencephalograph assessment aid.
(a) Identification. The neuropsychiatric interpretive electroencephalograph assessment aid is a prescription device that uses a patient's electroencephalograph (EEG) to provide an interpretation of the patient's neuropsychiatric condition. The neuropsychiatric interpretive EEG assessment aid is used only as an assessment aid for a medical condition for which there exists other valid methods of diagnosis.
(b) Classification. Class II (special controls). The special controls for this device are:
(1) The technical parameters of the device, hardware and software, must be fully characterized and must demonstrate a reasonable assurance of safety and effectiveness.
(i) Hardware specifications must be provided. Appropriate verification, validation, and hazard analysis must be performed.
(ii) Software, including any proprietary algorithm(s) used by the device to arrive at its interpretation of the patient's condition, must be described in detail in the software requirements specification and software design specification. Appropriate software verification, validation, and hazard analysis must be performed.
(2) The device parts that contact the patient must be demonstrated to be biocompatible.
(3) The device must be designed and tested for electrical safety, electromagnetic compatibility, thermal, and mechanical safety.
(4) Clinical performance testing must demonstrate the accuracy, precision, reproducibility, of determining the EEG-based interpretation, including any specified equivocal zones (cutoffs).
(5) Clinical performance testing must demonstrate the ability of the device to function as an assessment aid for the medical condition for which the device is indicated. Performance measures must demonstrate device performance characteristics per the intended use in the intended use environment. Performance measurements must include sensitivity, specificity, positive predictive value, and negative predictive value per the device intended use. Repeatability of measurements must be demonstrated using interclass correlation coefficients and illustrated by qualitative scatter plot(s).
(6) The device design must include safeguards to prevent use of the device as a stand-alone diagnostic.
(7) The labeling must include the following information:
(i) A warning that the device is not to be used as a stand-alone diagnostic.
(ii) A detailed summary of the clinical performance testing, including any adverse events and complications.
(iii) The qualifications and training requirements for device users including technicians and clinicians.
(iv) The intended use population and the intended use environment.
(v) Any instructions technicians should convey to patients regarding the collection of EEG data.
(vi) Information allowing clinicians to gauge clinical risk associated with integrating the EEG interpretive assessment aid into their diagnostic pathway.
(vii) Where appropriate, validated methods and instructions for reprocessing of any reusable components.
[79 FR 9085, Feb. 18, 2014]